From: Kenton E Graham (kgraham@CSWNET.COM)
Since I've had several requests regarding the mention I made on the wattmeter
that appears in "Joy of QRP", Here's some more info.... 73 de Ken K5ID
PS: a good 100 or 200 microamp meter works best....
>From: kgraham@CSWNET.COM (Kenton E Graham)
>Here's a "word picture" of the wattmeter...
>tie the anode of a 1n62 or ecg109 diode to the 50 ohms in your dummy load.
>The cathode end is grounded thru a .01 disk cap (100v). The cathode also
>goes to another enclosure with the meter in it. I use a bnc connection to the
>other box. In the other box the bnc goes to an rfc-- 1 mh, 300 microh, or
>8 turns on a ft3743, not critical-- then to resistor of about 200-300k to the
>plus side of the meter. (the resistor can be changed to give full scale at
whatever power level you want). Minus side of meter to ground.
>That is the complete circuit. If you remove the 50ohm resistor, the
>wattmeter can be calibrated as follows:Use an adjustable power supply and a
>apply the voltage and meter where the 50 ohms was. correlate lots of
voltage points with meter readings. 10 volts is about 5 watts, 22 volts is
about 5 watts.
>Using the formula P=Vpk*Vpk/2R you will get power. So 10 volts will represent
>1 watt if you have exactly 50 ohms.... etc.
>The diode renders your dummy load useless for high power unless you
disconnect the diode with a switch or something, because the back voltage is
about 60v for the ecg109 and about 100v for the 1n62. Don't forget, or you
will blow the diode!!
>Of course, this is not an inline wattmeter; but it is very handy in setting
up or determining power levels offline..
> good luck and 73, Ken, K5ID
Search QRP-L Archives
QRP-L Archive |
[ 1993 | 1994 | 1995 | 1996 | 1997 | 1998 | 1999 | 2000 ]
This archive was generated by hypermail 2b29 on Fri Jun 02 2000 - 11:30:51 EDT